HPC for Quanti-Tray
Heterotrophic plate count made easy for bottling and manufacturing plants and on-premise testing
- Quantifies heterotrophic organisms in a 100 mL format
- No autoclaving, filtering, or agar
- No colony counting or interpretation
- Convenient 100-mL sample size—test source water, transport vehicle water, manufacturing line water, and finished products.
- Simple procedure needs no special training.
- Shares the same Quanti-Tray system as Colilert, Enterolert, and Pseudalert tests for maximum efficiency.
- Results valid at 44–72 hours.
How the HPC for Quanti-Tray test works
The HPC for Quanti-Tray test is used for the quantification of heterotrophic organisms in 100 mL of water.
The test is based on the IDEXX proprietary Multiple Enzyme Technology, which detects viable heterotrophic organisms by testing for the presence of key enzymes known to be present in these organisms. The test uses multiple enzyme substrates that produce blue fluorescence when metabolised by waterborne bacteria.
The sample/reagent mixture is added to a Quanti-Tray, incubated, and then examined for fluorescing wells. The number of fluorescing wells corresponds to a most probable number (MPN) of total heterotrophic organisms in the original sample.
The HPC for Quanti-Tray test detects organisms at 1 cfu/100 mL after 44–72 hours of incubation.
How to use
Learn how to use the HPC for Quanti-Tray test
Add reagent to sample (sample will turn pink).
Seal using a Quanti-Tray Sealer; incubate for 44 hours.
Read under ultraviolet light; count blue fluorescent wells and refer to MPN table.
Frequently asked questions
The HPC for Quanti-Tray test is designed specifically for bottled-water samples. Bottled-water samples are defined as samples taken from any point in the bottled-water manufacturing process (from the original water source, to transportation vehicles, to the manufacturing line, to the finished product).
There is no comparator for the HPC for Quanti-Tray test. Use a negative control for comparison when you interpret results.
Yes, if the results are positive. An inoculated HPC for Quanti-Tray test sample that is positive before 44 hours is a confirmed positive test. However, for an accurate quantitative determination of heterotrophic count in a sample, the results should be read between 44 and 72 hours of incubation.
Yes, if the results are negative. If an inoculated HPC for Quanti-Tray test sample is inadvertently incubated over 72 hours, lack of fluorescence is a valid negative test. However, a fluorescent read after the 72 hours is not valid and should be repeated or verified.
Yes, both the Quanti-Tray and Quanti-Tray/2000 can be used with the HPC for Quanti-Tray test in the 100-mL format.
The HPC for Quanti-Tray test reagent powder should be pink in colour and of a free-flowing, granular consistency. Direct any concerns about the colour or integrity of the powder to IDEXX Technical Service at 1300 44 33 99.
Check with your local authority for proper disposal of bacteriological biohazard materials at your facility.
Resources & Tools
Tests & Accessories
Search the Water Tests & Accessories tool
HPC for Quanti-Tray Test (20 test pack)
Catalogue number: WHQT-20
HPC for Quanti-Tray Test (200 test pack)
Catalogue number: WHQT-200
HPC for Quanti-Tray /2000 Test (20-test pack combo)
Catalogue number: WHQT2K-20
HPC for Quanti-Tray /2000 Test (200-test pack combo)
Catalogue number: WHQT2K-200
IDEXX Water has reference materials and approval documents to support the many products in our water portfolio. Find the document(s) you need by selecting the link below.
Search the Reference & Regulatory Documents tool
Water Customer Support
Australia & New Zealand Head Office
IDEXX Laboratories Pty Ltd.
Unit 6, 38-46
Rydalmere, NSW 2116
Phone (AUS): 1300 44 33 99
Phone (NZ): 0800 838 522
Fax (AU): 1800 634 409
Fax: (NZ): 0800 448 443